Find great deals for HP a Spectrum Analyzer Display Mainframe W/ HP b Pattern Generator. Shop with confidence on eBay!. Find great deals for Hewlett Packard HP a Display With b Optical Spectrum Analyzer. Shop with confidence on eBay!. Software and documentation supplied. This service guide is part of an Option OB3 package which includes: HP A Service Guide. HP A Component.
|Published (Last):||8 September 2012|
|PDF File Size:||1.19 Mb|
|ePub File Size:||19.8 Mb|
|Price:||Free* [*Free Regsitration Required]|
Can you tell one from the other? Knowing I had nothing to loose, I went and opened the transformer can with a Dremel, and removed the resin with hot air and tweezers, carefully to not damage the windings. Something was drawing more current than expected and the current-limit protection was tripping. The block diagram was invaluable, this switching power supplies are complex.
Do you see that suspicious potting that squeezed outside the can in one of them?:. I was happy to find it actually worked I bought this cheap A color display as faulty.
This last one required some improvisation because it is a quad-transistor instead of dual, and the DIP package is very different.
I was lucky to have bought a FLUKE D differential voltage probe this year at Friedrichshafen to the nice guy with grey hair and glasses that sells mostly Fluke stuff.
You really need to know what you are doing before probing and testing a thing like this live.
The supply operates a high 700004a frequency and many parts of the circuit are shorted at DC, so I could not find this until I took both transformers out. High voltage, big and the troubleshooting will need to be done live. And a final picture of the twin transformers in the working power supply.
HP a Spectrum Analyzer Display Mainframe W/ HP b Pattern Generator | eBay
After many hours of testing and probing, and and few components that exploded in the process most times my fault I found there was a transformer with a 14 ohm leak between primary and secondary. I used a thin teflon tube and a copper wire to recreate the four turns that form the secondary and guess what, it worked!! Home Projects Music Photo. This is the beast on the bench, that PCB looks very timidating to me.
This paid for itself in just this repair. This board is a latter revision than the ones that appear on the schematics but there are not many changes.
Keysight Technologies (Agilent HP) A used or new for sale at used-line
I could see the power LED flash briefly upon turn-on and then I confessed before that I killed a few components on the process, and I could not find replacements for some of them: Could this be fixed?
There is a current sense resistor and this is what current looked at turn-on try:. You can see the power supply slowly increasing the current at turn on until it gets too high before it starts regulating the voltage, and quits.
No surpise, the secondary jacket was burned and touching the primary in one spot:. One day, it refused to turn on. I built this “adapter” thing as a temporary solution, you too know what that means: Many suggested changing every capacitor in the board, something I was not going to do, I wanted to find what the fault really was all capacitors turned to be ok:.
You’ll probably want to use an isolation transformer and differential probes, and keep one hand on your pocket at all times. 70004w did not know where to start. There is no way to check one of the blocks of the power supply alone, you have to give it power and hope for the best.